TECPEL - Propel Quality Measurement
This pocket meter is equipped with test lead storage holder.
Open the lid on back of pocket multimeter,
and place the test leads into the cavity for pack-up.
It's convenient to carry it in your pocket.
DMM-113B + NCV (Non Contact Voltage indicator)
Display: 3 1/2 digit liquid crystal display (LCD)
with a maximum reading of 3400,
with a 34-segment analog bar-graph.
|Model : DMM-113B|
Display: 3 1/2 digit liquid crystal display (LCD) with a maximum reading of 3400, with a 34-segment
Polarity : Automatic, (- ) Negative polarity indication
Overrange : " OL" mark indication
Low Battery indication: The " Battery sign" is displayed when the battery voltage drops below the
Measurement rate: 2 times per second, nominal.
Bar Graph: 12 times per second
Auto Power Off: Approx. 8 mininutes
Operating environment: 0 ~ 40 degree C ( 32 to 104 degree F)
Storage temperature : -20 deg C ~ 60 deg C ( -4 to 140 deg F) at < 80 % RH with battery removed
Temperature coefficient: 0.1 x ( specified accuracy) per degree C ( 0 ~ 18 C, 28 C to 40 C)
Altitude : 6561.7 Feet ( 2000 m)
Power : Two 1.5 button type batteries ( IEC # LR-44, NEDA # 1166A)
Dimensions: 125 mm (H)x 60 mm (W) x 24mm (D)
Weight: Approx. 3.9 Oz (110g) including batteries and case